Mitek VAPR ll Electrode System
The Mitek VAPR II Electrode System is designed to provide soft tissue ablation (vaporization), contouring, cutting and hemostasis of blood vessels during arthroscopic surgical procedures.
Versatile portfolio of VAPR electrodes – Choice of optimized electrode shape and performance features for different anatomical areas and targeted arthroscopic procedures
“Safety-Fault” feature – Protection of arthroscopic equipment from damage caused by proximity of radiofrequency energy
Automatic electrode ID detection for simple default setups – Optimized and safe performance for various electrode types without the need to manually adjust the settings
Individual control over ablation and coagulation – Flexibility and enhanced safety by adjusted ablation speed and depth of tissue effect based on anatomical area and individual patient factors
Mitek VAPR ll Electrode System Specifications
- Bipolar Mode:
- Blended Vaporization
- Power Output: Digital Controls
- Excessive Current Alarm: Yes
- System Failure Alarm: Yes
- Foot-switches: Twin electrical
- Hand-switches: None
- Visual Output Indicator: Yes
- Auditory Output Indicator: Yes
- Volume control: Yes, Digital
How Does the Mitek VAPR II Electrode System Work?
Vaporization Modes (V1, V2 & V3): The increasing mode number indicates that higher maximum voltages and power outputs are available to produce a more vigorous effect. On activation, electrosurgical current heats the saline to establish a vapor pocket around the active electrode (see Photo 1). Arcs within the vapor pocket produce local vaporization of tissue. The default setting for each electrode type is designed
to deliver the minimum power necessary to sustain the vapor pocket and produce the desired tissue effect.
Desiccation Mode (DES): In this mode, the lower power settings and lower voltages across the electrode are insufficient to produce a vapor pocket. The active current is delivered to the tissue to dehydrate and coagulate without sparking or vaporization. The current then flows to the return electrode through the saline irrigant.
Blended Vaporization Modes (BV1 & BV2): Again, the increasing mode number indicates a more vigorous effect. These modes provide tissue vaporization combined with homeostasis. In BV1 mode, the output switches between the V2 and DES modes and between V3 and DES in BV2 mode.
Mitek VAPR ll Electrode System Applications
- Ablation and excision of soft tissue and hemostasis of blood vessels
- Arthroscopic surgery of
MeniscectomyLateral ReleaseChondroplastySynovectomyACL DebridementPlica RemovalMeniscal Cystectomy
Labral Tear ResectionSynovectomyExcision of Scar TissueAcromioplastyBursectomySubacromial DecompressionChondroplasty
Fracture DebridementExcision of Scar TissueSynovectomyChondroplasty
SynovectomyCartilage DebridementFracture Debridement
Footswitch is included for the Mitek VAPR II Electrode System.
There are no reviews yet.